Find answers to all your questions. We receive requests daily. This intuitive questionnaire updated on a regular basis will help you out when in trouble!
The mp6 micropumps achieve a maximum flow rate of 7 ml/min and a pressure of 600 mbar at 100 Hz, 250 Vpp and SRS signal. Our mp6-AIR micropump achieves a flow rate of minimum 20 ml/min and a pressure of 100 mbar at 300 Hz, 250 Vpp and SRS signal. The flow rate of the […]
The micropumps from Bartels Mikrotechnik GmbH are suitable for pumping liquids and gases for a variety of different applications. For some applications it is important to have a sound level as low as possible. Bartels has measured the mp6 and the mp6-AIR micropump under different circumstances. For further information and detailed results about our micropumps please […]
Bartels Mikrotechnik has different electronics available in order to evaluate the micropumps. For laboratory applications, there is the extended micropump control mp-x. It is able to drive one micropump mp6, mp6‑pp or mp6-AIR or two micropumps mp5 with different driving signals, the full range of amplitude and frequency manipulation and has an USB interface. For […]
Driving the Bartels mp6 micropump with the controller mp-x at 100 Hz with 250 V, flow rates of typ. 7 ml/min and backpressure of typ. 600 mbar can be achieved. The flow rate of the pump is a linear dependency on the pressure. At 0 mbar pressure the maximum pump rate can be achieved and at the […]
In medical technology, devices are mostly connected to each other via Luer or Luer-Lock connectors. The Luer system is a standardized connection system for those systems. The customary used terms are “male – female” or “m – f”. While the simple Luer connection is comparable to a standard grounding plug (m) and a standard […]